001 WHAT'S IN THE BOX 002 OVERVIEW 01 Volume Control 02 Detachable Windshield foam 03 Voice Focus Boom…
This instruction manual is part and parcel of the product and must accompany every product sold.
All rights reserved, including translation.
Congratulations! You are now owner of a Vermeiren rollator!
Thank you for your trust in the products of Vermeiren. To support you on the use of this rollator and its operating options, this manual is offered. Please read it carefully; it will help you to get familiar with the operation, capabilities, and limitations of your rollator.
If you still have questions after reading this manual, do not hesitate to contact your specialist dealer. He/she will be glad to help you.
To ensure your safety and to prolong the lifetime of your product, please take good care of it and have it checked and serviced on a regular basis.
This manual reflects the latest product developments. Vermeiren has the right to implement changes to this type of product without any obligation to adapt or replace similar products previously delivered.
Pictures of the product are used to clarify the instructions in this manual. Details of the depicted product may deviate from your product.
On our website, http://www.vermeiren.com/, you will always find the most recent version of the information in this manual. Please consult this website regularly for possible updates.
Visually impaired people can download the electronic version of this manual and have it read out by means of a text-to-speech software application.
3.1 Upon delivery
Unpack your product and verify if the delivery is complete. The following items should be included:
Verify your product for transport damage. If you find any damages after delivery, contact the transporter.
Indications and contraindications: The rollator is meant as support of the leg function during walking. Do not use the rollator if you have other physical limitations that may cause unsafe use of your rollator, like a weak arm, balance disorders, impaired vision, limb loss, …
The rollator is designed and produced solely to be used as walking aid for one person with a maximum weight of 200 kg.
The rollator will compensate the lack of adequate support by the legs by leaning on, or moving with, the rollator with usage of the upper limbs or chest. The rollator will provide better stability, support and safety during walking.
The rollator can be used indoors and outdoors.
It is not intended for transport of goods and/or more people than the user.
It should only be used on surfaces where all four wheels are touching the ground. Do not use the rollator on rocky, bumpy or sandy terrain, or places with a steep inclination. The rollator may become unstable. Look for flat, stable areas, and bevelled kerbs to walk with your rollator.
The warranty on this product is based on normal use and maintenance as described in this manual. Damage to your product caused by improper use or lack of maintenance will cause the warranty to lapse.
Please check all technical details and limits of your rollator in §7.
CAUTION: Risk of injuries
Please read and follow the instructions in this manual. Otherwise you may get injured, or your rollator may get damaged.
Symbols on the rollator
Shipping and storage
The shipping and storage of the rollator should happen according to the technical specifications in §7. Make sure that the rollator is stored dry.
Provide sufficient covering or packaging to protect the rollator from rust and foreign bodies (e.g. salt water, sea air, sand, dust).
Make sure no objects are stored on or against the rollator to prevent damage.
CAUTION: Risk of injuries
Assemble the rollator completely, before adjusting its height.
CAUTION: Risk of falling or CAUTION tipping over Always set the four wheels to the same height.
When mounting the rear wheels, check that the brake cable connection (4) is positioned to the inside of the rollator.
Mounting the handgrips
To prevent the cable from being trapped between moving parts of the rollator, secure the brake cable to the frame using the cable clamp (7).
Mounting the storage bag
CAUTION: Risk of falling or tipping over
Do not overload the storage bag.
The storage bag is meant to be mounted below the seat, on the two horizontal tubes on which the seat rests.
Mounting/Removing the backrest
CAUTION: Risk of damage
Pay attention to the position of the brake cables. Don’t pull the handgrips too high to prevent damage to the brake cable.
The disassembly of the rollator is conducted in reverse order to the assembly:
CAUTION: Risk of falling or tipping over Always set the four wheels to the same height.
To adjust either of these:
Make sure that
The force of the brakes can be adjusted by loosening/tightening the screw (14) on the back of the brake lever.
CAUTION: Risk of falling.
Check the working of the brakes to ensure your safety. The condition of the brakes is influenced by wear, oil and other contamination of the rear wheels and braking mechanism.
The rollator is equipped with brakes, operated by two brake levers (15). The brake levers operate in two modes: running brakes and parking brakes.
CAUTION: Risk of damage Do not push the rollator too hard while using the running brakes. This may cause damage to the tyres or brakes.
CAUTION: Risk of entrapment Keep fingers, buckles, jewellery and clothes away from the folding mechanism.
CAUTION: Risk of falling Check that the rollator is completely open and that the frame is locked.
CAUTION: Risk of falling Take kerbs straight, with both front wheels aligned to the kerb. Keep the rollator nearby.
To make it easier to take small obstacles, such as kerbs, the rollator is equipped with a tipping aid on the two rear wheels.
Sitting down, standing up and walking
CAUTION: Risk of falling Make sure that both brake levers are in parking position before sitting down, see §5.1. Keep the brake levers in this position until you start walking again. Keep your feet between the rear wheels.
Regular care ensures that your rollator is preserved in a perfectly functional condition. For the maintenance manual, refer to the Vermeiren website: www.vermeiren.com.
Use a soft, damp cloth and lukewarm water with a mild household soap to clean the rollator. Avoid soaking.
Do not use abrasive cleaning agents to clean.
CAUTION: Risk of damage Disinfections may only be undertaken by trained persons. Consult your specialist dealer.
Use a soft, damp cloth and lukewarm water with a mild household disinfectant to clean the rollator.
The expected service life of your rollator is 5 years. The lifespan of the rollator is influenced by its use, storage, maintenance, service and cleaning.
Before you resume use of the rollator, it should be inspected on the following points:
Before each reuse, have the rollator disinfected, inspected and serviced according to the instructions in §6.2 and §6.3 .
6.5 Disposal at end of life
At end of life, you need to dispose your rollator according to the local environmental legislation. The best way to do so, is to disassemble the rollator to facilitate the transport of recyclable parts.
The technical details below are only valid for this rollator, at standard settings and optimal ambient conditions. Take these details into account during use.
|Maximum user weight||200 kg|
|Maximum weight of storage bag load||5 kg|
|Overall length||750 mm|
|Folded length||400 mm|
|Overall width||650 mm|
|Folded width||650 mm|
|Overall height||830 – 1020 mm|
|Handgrip height||830 – 1020 mm|
|Recommended body height||1650 – 2050 mm|
|Distance between handgrips||580 mm|
|Seat height||490 – 560 mm|
|Seat width||470 mm|
|Seat depth||410 mm|
|Total mass |
– without basket
|10,7 kg |
|Brakes||Running & parking brakes|
|Turning width||1050 mm|
|Max. inclination forwards||15°|
|Max. inclination backwards||7°|
|Max. inclination sideways||3,5°|
|Storage temperature||+5°C – +41°C|
|Storage humidity||30% – 70%|
|We reserve the right to introduce technical changes. Measurement tolerance +/- 15 mm / 1,5 kg / 1,5°.|
|VERMEIREN Goliat Sturdy Foldable Rollator Walker up to 200 KG [pdf] User Manual|
Goliat Sturdy Foldable Rollator Walker up to 200 KG
P796 18V Cordless 3-Tool Lighting Kit OPERATOR'S MANUAL MANUEL D'UTILISATION MANUAL DEL OPERADOR 18 VOLT…
NatureChlorMax Electrolyzed Water 12V 2A Power Supply NatureChlor NatureChlorMax Operating Instructions 12146 Charles Drive, Grass…
Foundation 3 to 5 Tons Packaged Rooftop Units Cooling and Gas/Electric Quick Reference Guide FoundationTM…
Z-Wave 16A Touch Termostat INSTRUCTION MANUAL NAMRON TERMOSTAT TOUCH Z- WAVE 16A INSTRUCTION MANUAL Namron…