ThermoWorks WAND Blue THD2FE-B Instruction Manual
Indications for Use
ThermoWorks WAND Blue is a non-contact digital forehead thermometer intended for the intermittent measurement of human body temperature in people of all ages.
Operating Instructions for Forehead Use
Press to turn on the instrument. Hold the instrument within 1.5 inches from the center of the forehead and press and hold `Start’ until you hear one beep (unless muted) to indicate the reading is complete. Readings should take approximately 1 second.
The instrument will automatically turn off after 60 seconds. Manually turn the instrument off by pressing and holding the for 5 seconds.
The temperature displayed is the “oral equivalent” value. If the temperature is above 99.5°F (37.5°C) you will hear three short beeps followed by one long beep.
For Best Results:
Keep the forehead area clean. Keep away from sweat, cosmetics, and scars while taking measurements.
Please remain in a stable environment for 5 minutes and avoid exercising, bathing, or showering for 30 minutes, which can artificially raise your temperature.
Bluetooth Instrument Operation
Power on WAND Blue and it will begin broadcasting a Bluetooth signal. Open app and connect to the instrument. Once connected, pressing the Start button will send temperature data.
Forehead Range — 93.2 to 108°F (34 to 42.2°C)
Surface Range — –7.6 to 176°F (22 to 80°C)
Operating Range — 50 to 104°F (10 to 40°C) 15 to 85% RH
Storage Range— – 4 to 122 (20 to 50°C), RH85%
Transport. Temp — Shall be less than 158°F (70°C), RH≤95%
Atmospheric Pressure — 800 to 1013 hp
Forehead Accuracy — ±0.4°F (0.2°C) within 95 to 107.6°F (35 to 42°C) Otherwise ±0.5°F (0.3°C)
Surface Accuracy — ±0.5°F (±0.3°C) with 17.6 to 108°F (22 to 42.2°C) Otherwise ±4% or ±4°F (±2°C), whichever is greater
Resolution — 0.1° Units °F/°C
Response Time — 1 second
Wireless — Bluetooth 4.2
Wireless Approval — US (FCC), Canada (IC)
Auto-Off — 60 seconds
Battery — 2 x AAA, 2,000 hours
Dimensions — 6.2 L x 1.9 W x 1.6 D inches (158 L x 48 W x 40.2 D mm) Weight — 3.5 oz. (100 g)
Expected Service Life — 4 years
Contents — Thermometer x 1, Operating Instructions x 1, Battery AAA x 2
When the “Low Battery” icon flashes, the batteries should be replaced immediately with 2x AAA, 1.5 volt batteries. To open the battery cover, use your thumb to press down and slide the battery cover off. Remove and properly dispose the used batteries promptly, keeping away from children. Insert the new batteries according to the correct polarity. Replace the battery cover.
Intended Operator – Have eight years of education, no maximum.
The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any use of such marks by ThermoWorks, Inc. is under license.
Storage & Cleaning
The instrument should be stored at room temperature, away from liquids and direct sunlight. If there are any temperature differences between the place where the instrument is stored and where you are going to measure, please allow the instrument to stabilize to the ambient temperature for at least 15 minutes. Holding the instrument too long may cause a higher ambient temperature reading. This could make the body temperature measurement lower than usual.
The sensor lens is the most delicate part of the instrument and should be kept clean at all times. Care should be taken when cleaning the lens to avoid damage. Use only a soft cloth or cotton swab with medical alcohol, allowing the lens to dry fully before using the instrument again.
This instrument carries a one-year warranty against defects in either components or workmanship. During this period, products that prove to be defective will, at the discretion of ThermoWorks, be either repaired or replaced without charge. Full details of liability are available within ThermoWorks Terms & Conditions of Sale at www.thermoworks.com/productwarranty.
For warranty, service, and technical assistance, please contact ThermoWorks’ Technical Support at (801) 756-7705 or email at [email protected]
This device complies with part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) this device may not cause harmful interference, and (2) this device must accept any interference received, including interference that may cause undesired operation.
This equipment complies with the FCC RF radiation exposure limits set forth for an uncontrolled environment. This equipment should be installed and operated with a minimum distance of 20 centimeters between the radiator and your body.
This device complies with Industry Canada License-exempt RSSs. Operation is subject to the following two conditions: (1) this device may not cause harmful interference, and (2) this device must accept any interference, including interference that may cause undesired operation of the device.
Contain FCC ID: SH6MDBT40 / Contain IC: 8017A-MDBT40
This is not as AP or APG product.
There is no gender or age limitation for using the infrared thermometer.
This device should not submerge into any liquids and expose it to direct moisture.
Choking from swallowing small parts and batteries by children or pets is possible, please keep small parts and batteries at places where children and pets can’t reach.
Changes or modifications not expressly approved by the party responsible for compliance could void the user’s authority to operate the equipment.
NOTE: This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the FCC Rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures:
- Reorient or relocate the receiving antenna.
- Increase the separation between the equipment and receiver.
- Connect the equipment into an outlet on a circuit different from that to which the receiver is connected.
- Consult the dealer or an experienced radio/ TV technician for help.
Documents / Resources
|ThermoWorks WAND Blue THD2FE-B [pdf] Instruction Manual|
WAND Blue, ThermoWorks, Non-Contact, Digital, Forehead, Thermometer, Bluetooth, Wireless, THD2FE-B