KKMOON Air Quality Monitor Instruction Manual
Display method: 4.3” LCD screen display, 480 x 270 pixels
Atmospheric pressure: 86Kpa – 106Kpa
Detection method for CO2 : Infrared (NDIR)
Sampling time: 1.5 seconds
Product Size: 145 x 78 x 97.2mm
Detection temperature: -10°C to 50°C;
Relative humidity : 20% – 85%
Storage temperature: -10°C to 60°C;
Concentration unit for CO2: PPM
Power source: Lithium battery with 3000 mAh capacity; 5V DC power charging via micro USB port
Product weight: 210g
This product is an air quality monitor that detects Carbon dioxide (CO2),Temperature, and Humidity. it combines sensors real-time monitoring of Carbon dioxide (CO2), temperature, and humidity on its digital LCD display.
Please read the instructions carefully before using this device.
Please let the device work short mins outdoors before use for most accurate results .
Please keep the manual handy for quick reference and troubleshooting.
Avoid covering the air intake areas during use to avoid inaccurate measurements.
Avoid use of solvents to clean the product as residual fumes will skew air quality readings.
Avoid water or other liquids near the product to avoid electrical damage.
Do not allow unauthorized modification or repair of this product.
- 4.3″ color liquid crystal display (LCD), 480 x 270 pixels
- Test variables:Carbon dioxide (CO2), temperature, humidity
- Large 3000mAh capacity Lithium battery
- 5V Micro USB charging
- Low battery warning
1. Start Up
When you long-press the center power button, the air quality monitor will boot up.
- Power ( ON/OFF ) / OK / Menu Button, used to confirm highlighted options or to turn device on/off by pressing for 3 seconds.
- Down Decrease Button.
- Up Increase Button.
- Micro USB charging Port.
2. Data Display Formats ( Figure 1)
3.CO2 data record (Figure 2)
Press the up or down button to navigate to the trend screen. Graph shows the last 5 data values for Carbon dioxide taken every 30 minutes over the past 2.5 hours.
4.Set up (Figure 3)
Double press on/off button to enter the Set up screen.
DATE: Change the date by using the up and down button. Confirm each setting with one press on the on/off button.
TIME: Change the time by using the up and down button and by confirming with the on/off button.
UNIT: Change the temperature unit between Celcius (℃) or Fahrenheit (℉).
TEMP: Enable a temperature offset.
5. About Charging
When low battery icon is displayed, the device needs to be charged. Insert the included or another compatible micro USB charging cable into the device. Attach the other end to a USB DC charger (such as a smart phone charger) that outputs DC 5V at >=1000mA. Fully charge for at least 2-3 hours before use. Avoid charging with a USB computer port which only outputs 500mA.
CO2 technical indicators
Measuring range: 400-5000PPM
Resolution: 1 PPM
Sensor for CO2 : Infrared ( NDIR)
Temperature and humidity technical indicators
Measuring range: -10-50 ° C
Measurement accuracy: ±1 °C
Humidity range: 20%-85% RH
Measurement accuracy: ±4% RH
Air Quality monitor x 1
Micro USB charging cable x 1
Product Manual x 1