BraunAbility Transfer Board Overview How to Use: Risk for seat belt damage: Make sure that the seat belt…
This user manual contains important information about the handling of the product. To ensure safety when using the product, read the user manual carefully and follow the safety instructions. Invacare reserves the right to alter product specifications without further notice. Before reading this document, make sure you have the latest version. You find the latest version as a PDF on the Invacare website. If you find that the font size in the printed document is difficult to read, you can download the PDF version from the website. The PDF can then be scaled on screen to a font size that is more comfortable for you. In case of a serious incident with the product, you should inform the manufacturer and the competent authority in your country.
We provide a manufacturer’s warranty for the product in accordance with our General Terms and Conditions of Business in the respective countries. Warranty claims can only be made through the provider from whom the product was obtained.
within the disciplines of ISO 13485. This product features the CE mark, in compliance with the Medical Device Regulation 2017/745 Class I. This product features the UKCA mark, in compliance with Part II UK MDR 2002 (as amended) Class I. We are continuously working towards ensuring that the company’s impact on the environment, locally and globally, is reduced to a minimum. We only use REACH compliant materials and components.
The product has been tested and conforms to the standard ISO 17966 (Assistive products for personal hygiene that support users) and related standards. For further information about local standards and regulations, contact your local Invacare representative. See addresses at the end of this document.
Symbols in this Document
Symbols and signal words are used in this document and apply to hazards or unsafe practices which could result in personal injury or property damage. See the information below for definitions of the signal words.
Indicates a hazardous situation that could result in serious injury or death if it is not avoided.
Indicates a hazardous situation that could result in minor or slight injury if it is not avoided.
Indicates a hazardous situation that could result in damage to property if it is not avoided. Tips and Recommendations Gives useful tips, recommendations and information for efficient, trouble-free use.
UK Responsible Person Indicates if a product is not manufactured in the UK. Triman Indicates recycling and sorting rules (only relevant for France).
The expected service life of this product is three years when used daily and in accordance with the safety instructions, maintenance intervals and correct use, stated in this manual. The effective service life can vary according to frequency and intensity of use.
Limitation of Liability
Invacare accepts no liability for damage arising from
WARNING! Risk of Serious Injury or Damage
WARNING! Risk of Falling
CAUTION! Risk of Injury
Labels and Symbols on the Product
The identification label is attached to the underside of the bath board.
Scope of Delivery
Invacare Marina is a bath board to be placed on the bathtub’s rim and intended to be used as a shower aid.
Intended users are any adults and adolescents with restricted or limited mobility. If used without an assistant, the users must have sufficient torso stability to keep their own balance.
There are no contraindications known when using the product as intended. For the maximum user weight of the product see 14 Technical Data
Assembling the Product
The bath board is suitable for bathtubs with an inner width of 400 to 620 mm
Attach the handle to the seat board in the correct orientation
Make sure that the handle fully engages in the seat board.
The handle can not be removed again after assembly.
Placing the Bath Board on the Bathtub
WARNING! Risk of injury
Using the Product
WARNING! Risk of Injury
General Maintenance Information
This product is maintenance-free, provided that the cleaning and disinfection instructions are adhered to.
General Safety Information
Wrong fluids or methods can harm or damage the product
For cleaning and disinfection in clinical or long-term care environments, follow your in-house procedures.
Cleaning in a Clinical Environment
In addition to cleaning as described above we recommend cleaning the product prior to the use by another patient, after ending of the period of use or on a regular basis (depending on which occurs first) at
Be environmentally responsible and recycle this product through your recycling facility at its end of life. Disassemble the product and its components, so the different materials can be separated and recycled individually. The disposal and recycling of used products and packaging must comply with the laws and regulations for waste handling in each country. Contact your local waste management company for information.
This product is suitable for reuse. To recondition the product for a new user, carry out the following actions
For detailed information, see 11 Maintenance, page 4 . Make sure that the user manual is handed over with the product. If any damage or malfunction is detected, do not reuse the product.
Invacare Australia Pty. Ltd. Unit 18/12 Stanton Road, Seven Hills, NSW 2147, Australia Phone: 1800 460 460 Fax: 1800 814 367 firstname.lastname@example.org www.invacare.com.au
Invacare New Zealand Ltd 4 Westfield Place, Mt Wellington 1060 New Zealand Phone: 0800 468 222 Fax: 0800 807 788 email@example.com www.invacare.co.nz
Invacare Poirier SAS Route de St Roch F-37230 Fondettes Tel: (33) (0)2 47 62 64 66 firstname.lastname@example.org www.invacare.fr
Invacare Limited Pencoed Technology Park, Pencoed Bridgend CF35 5AQ Tel: (44) (0) 1656 776 200 email@example.com www.invacare.co.uk
Invacare Canada L.P. 570 Matheson Blvd East, Unit 8 CDN Mississauga, On. L4Z 4G4 Phone: (905) 890 8300 Toll Free: 800.668.5324 www.invacare.ca
Invacare GmbH Am Achener Hof 8 D-88316 Isny Germany Invacare UK Operations Limited Unit 4, Pencoed Technology Park, Pencoed Bridgend CF35 5AQ UK
|INVACARE H112 Marina Bath Board [pdf] User Manual|
H112 Marina Bath Board, H112, Marina Bath Board, Bath Board, Board
SM 3775 Cordless Hand Blender ART.-NO. SM 3775 DE Gebrauchsanweisung GB Instructions for use FR…
REXR212-23 LED Flexi Strip RECESSED ALUMINIUM EXTRUSION 21.7MM DEEP 20.4MM INNER WIDTH 2M LONG REXR212-23…
Miele H 2467 recipes Oven User Manual
Big Farm Tractor with Wagon 8+ EN SAFETY PRECAUTIONS PLEASE READ! (Keep manual for…
JDF604E3 Cooker Hood %* &6 2GVDYDÏSDU 1£YRGNSRXLW¯ '$ (PK¨WWH %UXJVDQYLVQLQJ '( +HUG$E]XJVKDXEH %HQXW]HULQIRUPDWLRQ (/ (1…
FQC8222 Clixx Wireless Earbuds FLINQ CLIXX WIRELESS EARBUDS FQC8222/8223/8224/8225 Thanks for purchasing this FlinQ product!…